Influence of Irrigant Solutions and Apex Locators in Working Length Determination

instrumentation and obturation limit [1]. This mensuration is one of the main factors in endodontic treatment success [2]. Previous determination of working length is critical to cleaning, shaping and obturation of radicular canals and has to be carefully measured [3]. Influence of Irrigant Solutions and Apex Locators in Working Length Determination Reynoso RF1, Nazario RN2, Rodriguez J3, Bittencourt BF4, Dominguez JA5* 1Student, Department of Endodontics, Faculty of Health Science, Norbert Wiener University, Lima, Peru 2Professor, Department of Endodontics, Faculty of Health Science, Norbert Wiener University, Lima, Peru 3Professor, Sp Department of Endodontics, Faculty of Health Science, Norbert Wiener University, Lima, Peru 4Professor, PhD, Department of Dentistry, State University of Ponta Grossa, Ponta Grossa, Brazil 5Professor, PhD, Department of Endodontics, Faculty of Health Science, Norbert Wiener University, Lima, Peru


Journal of Oral Health and Dental Science Open Access
The use of electronic devices to determine the working length was first proposed by Custer (1918) [4], and the scientific basis of apex locators originated with the investigation by Suzuki, in 1942, that studied the current flow across the teeth of dogs.Consistent values in the electrical resistance between an instrument in a root canal and an electrode in the membrane of the oral mucosa was recorded, and may speculated that this would serve as a measure of the canal length [5].Sunada (1962) was the first author to describe the details of a simple clinical device for measuring the working length in patients.Since then, electronic apical locators have become an invaluable tool in the practice of modern endodontics [6].
Apex locators may provide more accurate estimation compared to radiographs [7].In a study which compared the conventional radiograph and the electronic apex locator method (Raypex 5), two blind evaluators concluded that Raypex 5 was superior than the conventional radiographs to determine the working length [7].
The use of irrigant solutions is an important step in endodontic treatment, once these products are fundamentally used to clean and disinfected the root canal [8].However, these solutions may affect the performance accuracy of apex locators.[9].evaluated three apex locators (Root ZX, Propex and Neoson EZ) and three irrigant solutions (0.9% NaCl, 3% H 2 O 2 , 2.5% NaOCl and 17% EDTA), and found decreased accuracy of Root ZX and Propex, due to the higher amount of electrolytes; being Neoson EZ the

Statistical Analysis
Scores (in %) were attributed to each difference found between the working length gave by the apex locators and the RWL.D`Agostino normality test was applied for the data, and t-paired test was used to analyze statistical differences between the RWL and the WL determined by the apex locators, for each experimental group.most accurate apex locator.[10].analyzed PROPEX after pulpal excision with three irrigants: 2.5% NaOCl, 0.9% NaCl and 0.2% chlorhexidine gel, demonstrating that the highest discrepancies occurred with 0.9% NaCl, and the lowest with 0.2% chlorhexidine gel.
Based on the abovementioned considerations, the objective of this study is to evaluate the accuracy performance of three apex locators using four irrigant solutions to determine the working length of root canals.
Radiographs were performed in mesiodistal and buccolingual views from all teeth.The specimens were glide path with a K10 file, and planned with a cilindrical diamond bur (MDT, Afula, Israel) in order to obtain a reference point for all the evaluations.

Specimen preparation
The endodontic access was conducted with a high speed spherical diamond bur (MDT, Afula, Israel).The cervical portion was enlarged with Gates-Glidden burs # 1, 2 and 3 (Dentsply Maillefer, Ballaigues, Switzerland).The roots were irrigated with 10 ml of saline solution and dried with absorbent papers.
After the working length mensuration with the apex locators, the real working length (RWL) was determined using a # 15 K-file (Dentsply Maillefer, Ballaigues, Switzerland) in each specimen until the end of the file was visualized in stereoscopic microscopy (Global, USA).This length was adjusted with a silicon endo stop; the distance between the silicon stop until the end of the file was calculated with a digital calibrator (Stainless Hardened, China).This procedure was repeated three times, and an arithmetic mean was made, determining the RWL.
The score distribution showed high differences (between 0.01 and 05) in all groups (Table 1).For the apex locators Raypex 6 and RootOR, the percentages were higher than 1.1, being statistically different from Endo EZE (between 0.51 and 1).
The mensurations made with EndoEZE showed that this device gives 100% of accuraccy ( ± 0.5 mm until the apex foramen), which may be considered an acceptable range [18].Also, when the canals were irrigated with 4% sodium hypochlorite, there was a slight difference compared to the other irrigant solutions, maybe because sodium hypochlorite has higher conductivity.These differences were not statistically significant, as the 0.5 mm range was clinically acceptable.
A recent meta-analysis evaluated the precision of apex locators in determining the RWL.The authors concluded that the device used and the type of irrigation influenced the results [19].In order to verify the precision of three apex locators, this study showed that the performance of EndoEZE apex locator was not influenced by the irrigant solutions to determine the working length.Similar results were found in the literature [11,12].The accuracy of RootOR and Raypex 6 was influenced by the use of the four irrigant solutions.[9].found that accuracy of Root ZX and Propex was decreased, as glass tubules were full of electrolytes.The main reason for the different results may be the parallel glass tubules used, without tapered or constriction, as teeth [9].In natural anatomy of a permanent tooth, the tubule walls are thicker along its length, and electric properties of glasses are different from human dentine.Also, Ozsezer et al. (10) concluded that Propex was more accurate when the root canal was full of Chlorhexidine, compared to NaOCl solution.This discrepancy may be attributed to the product concentration (0.2% rather than 2%); the 2% concentration used in this study is commonly cited in the literature as an effective irrigant solution [8].
The results of this work found there is a significant difference between the RWL and that one found with the apex locators Raypex 6 and RootOR.Saraswathi [20] found similar results with Raypex 5. Differently from our findings, other study showed high accuracy with Raypex 6 [21], even when compared with conventional radiographs [22,23].demonstrated that RootOR showed an accuracy of 86.7%.In this study, the authors measured the distance from the file tip until the major foramen under stereomicroscopy.Other study showed that RootOR was affected by 2.5% sodium hypochlorite (NaOCl), 0.9% saline solution (NaCl), and 17% EDTA.The authors concluded that better results could be found in dry canals [24].Endo EZE was the only apex locator that did not show differences with the 4 irrigant solutions.This fact may be due to Endo EZE locator reached the apical foramen and the operator should move back 0.5 mm, according to the manufacturer`s instructions, which ensures more security to determine the working length.As other authors reported, the operator should retrograde 0.5 to 1 mm from the indicated position by the apex locators, in order to establish the correct working length [13][14][15].

Table 1 :
Differences distribution (in %) between the working length determined by the apex locators and RWL, according to the irrigant solutions used.*